Easy return of products
This product is available in our stores
- vaginal probe PERISPHERA-AT
- user manual
thanks to which they have properties
BIOSTATIC AND BIOCOMPATIBLE, IDEAL FOR CONTACT WITH MUCUS MEMBRANES
PERISPHERA-AT is a vaginal probe for pelvic floor electrostimulation and EMG biofeedback. The probe is recommended for the treatment of urinary incontinence and in disorders in the intimate sphere. The probe is equipped with 4 gold-plated electrodes in nickel-free technology, making them ideal for people allergic to this element. Additionally, the presence of gold ensures good biostatic and biocompatible properties, therefore the probe is perfect for contact with mucous membranes.
The PERISPHERA-AT probe has 2 current circuits consisting of 4 electrodes (2 front and 2 back) that are positioned along the probe. Their purpose is to deliver current pulses to the vaginal mucosa. The current flowing between the electrodes creates a stimulus that affects the perivaginal muscles.
The electrode is equipped with a safe handle with a special collar that allows you to adjust the depth of insertion of the probe and the degree of rotation, as well as facilitating convenient application and removal of the probe. In addition, the comfort of use is influenced by the small diameter of the probe, which is only 11 mm, which is especially important for women with limited access to the vagina.
The 4 electrodes (2 on the front and 2 on the back) can be connected to the separate output channels of the pacing unit or to one channel using special optional Y-shaped wire adapters. The probe fits cables with a 2mm pin connector.
- small diameter of 11 mm
- comfortable handle with a collar for easy application and adjustment
- gold-plated electrodes with nickel-free technology
- latex free
- 2 channels
- 4 electrodes
- comfort of use
- class IIa medical device (93/42 / EEC)
CLEANING AND HYGIENE:
After each use, the probe should be washed with mild soap and water. If necessary, the probe can be disinfected with alcohol and rinsed with water before use. The probe should not be sterilized. The probe is intended for personal use. After 60 sessions, if there is a need for further use, the probe should be replaced with a new one.
- infection of the anus or rectum,
- cancer of the colon or the genitourinary system,
- pacemaker - consultation with a doctor is necessary
ATTENTION! For hygiene reasons, any return of the product under the 14-day right of return will not be considered if the probe is unpacked and the seal is removed from the packaging.
|Active surface of the electrodes:|
lenght: 12 cm,
diameter: 11 cm,
application length: 6 cm